Why is it that measurement data cannot be displayed on the PC by the PCA 3000 evaluation software, even though there was no error when loading the measurement data? Why does the measurement display on the Logoscreen nt screen go missing after you change the view on the Logoscreen nt?
View in the PCA 3000 evaluation software:
To enable the Logoscreen nt to store measurements in normal mode, the "Status" parameter must be changed from the factory setting of "Display only" to "Display, Save" in the particular group.
Apart from that, the storage status for normal mode must remain set at "ON".