What is to be taken into account for the error messages F022 and F023?
The error messages F022 and F023 indicate that the measuring range is gone below or exceeded.
Hold is additionally displayed. In this case, we recommend to check the correct actual value recording.
Check the measuring cell, cables, connectors and the set measuring range "ranG“ with the cell constant specified on the measuring cell.